The Agilent 1200 Series G1367B ( G1367 B / G 1367 B / G 1367B ) high performance Autosampler ( HPALS ) adds maximum flexibility and fast injection cycles, whenever high sample throughput and speed of analysis are required. The standard metering device (for the G1367B) provides injection volumes from 0.1–100 μl. A multi-draw kit extends the range up to 1500 μl.
System features include:
- Increased sample injection speed for high sample throughput
- Overlapped injections for increased productivity
- Minimal delay volumes for rapid gradients and fast equilibration when bypassing the autosampler after sample injection.
- Flexible and convenient sample handling with different types of sample containers.
- Using 384-well plates allows to process up to 768 samples unattended.
|GLP features||Early maintenance feedback (EMF), electronic records of maintenance and errors|
|Communications||Controller-area network (CAN). RS232C, APG-remote standard, optional four external contact closures and BCD vial number output|
|Safety features||Leak detection and safe leak handling, low voltages in maintenance areas, error detection and display|
|Injection range||0.1 – 100 µl in 0.1 µl increments Up to 1500 µl with multiple draw (hardware modification required)|
|Precision||Typically < 0.25% RSD from 5 – 100 µl, Typically < 1% RSD from 1 – 5 µl variable volume|
|Pressure range||400 bar (5880 psi)|
|Sample viscosity range||0.2 – 5 cp|
|Sample capacity||2 × well plates (MTP) + 10 × 2 ml vials 108 x 2-mL vials in2 x 54 vial plate plus 10 additional 2 mL vials 30 x 6-mL vials in 2 x 15 vial plate plus 10 additional 2 mL vials54 Eppendorf tubes (0.5/1.5/2.0mL) in 2 x 27 Eppendorf tube plate
Also compatible with the Agilent 1200 Series sample capacity extension for further expansion of the sample capacity
|Injection cycle time||Typically < 30 s using following standard conditions: Default draw speed: 200 µl/min Default eject speed: 200 µl/min Injection volume: 5 µl|
|Carry-over||Typically < 0.01% using the following conditions: Column: 125 x 4 mm Hypersil ODS, 5 µm Mobile phase: Water/Acetonitrile = 80/20 Flow rate: 1 ml/min Injection volume: 1 µl caffeine (1 mg/ml), 5 µl water to test carryover Outside wash of needle before injection: 20 sec with water using flush port|
Note: Licenses for Agilent´s software products are not transferable. Any customer who purchases Agilent products are advised to contact Agilent for license requirements.
All Brand Names and Logos depicted at our website are the trademarks of the respective companies. We are not their representative nor authorized resellers.
- Cables and connectors