- Cables and connectors
The specially-designed thermostatted sample trays hold either 100 × 1.8 ml vials or two well plates and 10 × 1.8 ml vials. The thermostatted air enters the autosampler through a recess underneath the specially-designed sample tray. The air is then distributed evenly through the sample tray ensuring effective temperature control, regardless of how many vials are in the tray. In cooling mode condensation is generated on the cooled side of the Peltier elements. This condensed water is safely guided into the leak system.
The Agilent 1200 ALS Thermostat, G1330 controls temperature for either HPLCs or fraction collectors. The ALS thermostat contains Peltier-controlled heat exchangers. A fan draws air from the area above the sample vial tray of the autosampler and is then blown through the fins of the cooling/heating module. There it is cooled or heated according to the temperature setting.
Note: Licenses for Agilent’s software products are not transferable. Any customer who purchases Agilent products are advised to contact Agilent for license requirements.
All Brand Names and Logos depicted at our website are the trademarks of the respective companies. We are not their representative nor authorized resellers.